On Jul 23, 2012 11:29 PM, "Kevin Reynolds" <tpresence at hotmail.com>
I have tried and failed to get a successful connection between my uvax II
and a pc
rs-232 serial port. The H8751-B didn't seem to work, so I pulled
it off and wired my own MMJ-DB9 connector. I have tested the MMJ cable,
and it works flawlessly.
If you have double checked a couple different cable options already, have
you double checked that the bulkhead panel for the KA630 is working
correctly? Is it possible that the battery leaked and corroded the PCB
traces on the bulkhead panel?
I've used both a homemade KA630 DB9 - PC DB9 cable and also an H8571-B MMJ
cable without any issues. It shouldn't be difficult if the KA630 panel is
working correctly and cabled to the KA630 correctly.